What are good applications for the new 95" boom?
I was thinking about using it on a standard 60-m tower instead of the 60" boom. Looking at a tower shadow sketch, I see that getting a sensor only four tower diameters away from the tower would give a reading of at worst +/- 2% of free stream. For a 10" tube, that's only 40", and the standard boom is 60".
So even if I were trying to get very clean readings at 10m AGL (on the 10" part of the tower), the 60" boom would be fine for every circumstance except for when the sensor is directly downwind (when you're screwed no matter what).