pcorcoran Posted November 1, 2011 Report Share Posted November 1, 2011 Although the Symphonie and SymphoniePLUS wiring panels both have the same db27 connectors, they are not to be used interchangeably. For example, do not wire a Symphonie classic with a SymphoniePLUS wiring panel and vice versa. The classic Symphonie (serial numbers starting with 3090xxxxx) have a 12 channel wiring panel that supports six counter channels and six analog channels. If a SymphoniePLUS wiring panel is used on a classic Symphonie, the data logged on counter channels 1 through 3 will be duplicated on channels 13 through 15, and the data may be distorted. The SymphoniePLUS (serial numbers starting with 4280xxxxx) has nine counter channels and six analog channels. If a classic Symphonie wiring panel is used on a SymphoniePLUS it may log data on the first 12 channels. If channels 13 through 15 are used the data will be significantly distorted. A SymphoniePLUS3 wiring panel will not fit on either of the older Symphonie loggers. Link to comment Share on other sites More sharing options...
This topic is now archived and is closed to further replies.